Using a large, high-definition LCD screen, FlexVision XL allows you to display multiple images in a variety of layouts — each tailored for your specific procedure. Personalization streamline interventions Everything from patient data management, exam scheduling, image acquisition, system movement, image post-processing and archiving — can be set to each clinicians specific work style. Intuitive controls at tableside simplify operations The Philips Allura system enables you to focus on the patient with a user interface that is intuitive and ergonomic, making system operation transparent. Easy patient access Together with its true iso-centric tilt and cradle capabilities, the Allura gives you flexibility in patient positioning.
|Published (Last):||3 January 2006|
|PDF File Size:||19.91 Mb|
|ePub File Size:||1.33 Mb|
|Price:||Free* [*Free Regsitration Required]|
Sep 26, PM :: 3 minute read :: The Philips Allura FD20 and FD10 may be similar in appearance at a glance, but they are very different machines in a number of ways.
The main difference between them is the work they are used for. The FD10 is primarily a cardiac system. It does have some options that may allow for light peripheral work, but cardiac studies are what it was designed for. The FD20 was designed, when equipped with the appropriate software, to handle a much broader spectrum of applications, including but not limited to : angio, cardiac, neuro, and vascular.
Ceiling Mount Mostly While both ceiling and floor-mounted gantries were manufactured for the FD10 and FD20, the vast majority are ceiling-mounted.
Floor-mounted FD10 units are hard to find, and floor-mounted FD20s are exceptionally rare. The Table There are two tables available for the FD series. It features a motorized, floating top with table-side controls.
AD7 tables have a tilting variant as well. User Interface Both the FD20 and FD10 have the same basic user interface: In the procedure room, the geometry module controls the motion of the table and gantry. The imaging module controls the X-ray generator, the tube, and the detector collimation.
The Xper module touchpad and controls available in later renditions allow for touchscreen and remote control of acquisition settings and imaging processing as well as optional modes of operation. The systems come standard with two LCD monitors on a ceiling-mounted boom that is capable of holding 4 to 8 monitors, depending on configuration and application. In the control room, there is a viewing console with one LCD monitor for viewing patient data and managing system information, a review monitor, and a review module that helps with exam viewing and post processing.
You can also have a second Xper module in the control room for added flexibility. The FD20 is geared to be slightly faster in movement and more flexible in positioning to accommodate angio studies.
These procedures need to cover a greater percentage of the body than those generally performed on the FD10, and the flexibility of the positioner is in keeping with those needs. The notable differences between these tubes are focal spot and heat dissipation. The focal spots of these tubes are different because of the size of the detectors they are paired with. The "" indicates its 0.
Detector The biggest difference between the detectors on these systems is their size, which, when measured diagonally, dictates the maximum field of view FOV each offers. Another difference between the detectors on these two systems is that the detector and collimator on an FD20 can rotate. If your day-to-day is more often comprised of cardiac cases, the FD10 would be a better pick. Contact us with any other questions! Out of the office, he spends most of his time keeping up with his wife and their three energetic sons.
Philips Allura XPER FD20/20 Cath & Angio
Sep 26, PM :: 3 minute read :: The Philips Allura FD20 and FD10 may be similar in appearance at a glance, but they are very different machines in a number of ways. The main difference between them is the work they are used for. The FD10 is primarily a cardiac system. It does have some options that may allow for light peripheral work, but cardiac studies are what it was designed for. The FD20 was designed, when equipped with the appropriate software, to handle a much broader spectrum of applications, including but not limited to : angio, cardiac, neuro, and vascular.
Philips Allura FD20