|Published (Last):||17 April 2011|
|PDF File Size:||1.67 Mb|
|ePub File Size:||7.89 Mb|
|Price:||Free* [*Free Regsitration Required]|
JoJorn Recommended Separation Distances This table details the recommended separation distances between portable and mobile RF communications equipment and this device. Model or M, field service by non-Nonin personnel, tampering, or any kind of misuse or abuse of the Model or M, shall void the warranty in its entirety.
Medical electrical equipment needs special precautions. This device contains WEEE materials; please contact your distributor.
Any sign or evidence of opening the system, field service by non-Nonin personnel, tampering, or any kind of misuse or abuse of the system, shall void the warranty in its entirety. Installing Batteries in the For more information about Nonin parts and accessories: Fluctuating or very bright light, moisture, blood pressure cuffs, infusion lines, venous pulsations, insufficient pulse signals.
Electromagnetic Immunity Technical Information Table 4: the Sensor Connect the pulse oximeter sensor with the Nonin logo facing up to the top of the device as shown. Comments to this Manuals Your Name. Nonin Onyx Instruction Manual Amplifier … AeroMedix is a global retailer of medical and instructlon products for the aviation industry. Indicates separate collection for electrical and electronic equipment WEEE. Nonin Onyx Finger Pulse Oximeter.
Page 23 Model or M, field service by non-Nonin personnel, tampering, or any kind of misuse or abuse of the Model or M, shall void the warranty in its entirety. Page 9 Do not attempt to open the case or repair the electronics. Due to the very small trace size, extreme care must be used when replacing components to prevent permanent, non-repairable damage to the circuit board.
Wrist-worn pulse oximeter with bluetooth wireless technology 51 pages. Page 4 Contents Continued Accuracy Testing Replacing the batteries before 3 minutes have elapsed most likely will result in corrupt data. They were all once published service manual for nonin onyx by Interplay!
Do not autoclave or immerse the device or sensors in liquid. Visual Indicators Visual Indicators The following table describes conditions and visual majual. A non-compatible sensor is being Replace the sensor with a used.
Communications Communications Serial Output Models and M provide real-time data output capability via the pulse oximeter sensor connector a 9-pin Sub-D connector.
Contact the carrier immediately if the shipping carton is damaged. General operation of the device may be affected by the use of an electrosurgical unit ESU.
Portable and mobile RF communications equipment should be used no closer to any part of the device, including cables, than the recommended separation distance calculated from the equation applicable to the frequency of the transmitter. Portable and mobile RF communications equipment can affect medical electrical equipment. Table of Contents Add insyruction my nojin Add.
For instructions on cleaning pulse oximeter sensors, refer to the respective sensor instructions for use. Related Articles.
9500 INSTRUCTION MANUAL NONIN PDF
It is intended for spot-checking of adult and pediatric patients on fingers other than the thumb between 0. Caution: Federal law USA restricts this device to sale by or on the order of a licensed practitioner. It must be used in conjunction with other methods of assessing clinical signs and symptoms. If adjacent or stacked use is necessary, the device should be observed carefully to verify normal operation. Significant levels of dysfunctional hemoglobin may affect measurement accuracy. Patient sensitivity to sensors may vary due to medical status or skin condition. Warm or rub the finger, or re-position the device.
Nonin Onyx 9500 Instructions For Use
Nonin recommends using alkaline batteries included with each new device. When batteries are low, the numeric displays flash once per second. Verify that all LEDs illuminate during the startup sequence. After the startup sequence, the device begins sensing the pulse indicated by the blinking pulse quality indicator. Dry with a soft cloth, or allow to air dry. Ensure that all surfaces are completely dry.
Do not use this device in an MRI environment, in an explosive atmosphere, or on infant or neonatal patients. Memory Functions m Only 3 minutes after removing the batteries. This device is intended only as an adjunct in patient assessment. This device should not be used adjacent to or stacked with other equipment. Opening the case may damage the device and void the warranty.
Nonin Onyx Vantage 9590 Instructions For Use Manual